Dr.Super 4 Water Spray Fiber Optic Dental Handpiece
----With quick-connect design, a key plug, easy installation and disinfection. The user's experience can not be better.
----Stainless steel push button with clear thread, avoid bur dropping
----Original bearing from Japan nsk, high tensile ceramic material and wear resisting
----Quick-connector bulbs for easy installation, an extra bulb accessory for extended life
----4-pot water spray better cooling effect, so that the patient is no longer uncomfortable
----Imported good fiber optic, good ductility
----Cartridge consist of dust cover, avoid cross injection and extend lifespan of the 4 water spray fiber optic dental handpiece
Operation Air Pressure
≥350(krpm)for Standard head; ≥280 (krpm) for torque head
LED Luminous Intensity
4 water spray
1. Closed cartridge protect of bearing and extend lifespan
2. Anti-retraction function design, protect patients from cross infection, extend lifespan for 4 water spray fiber optic dental handpiece
3. 135℃ autoclavable for the handpiece
4. Imported bearings, excellent internal quality and structure, resulting in outstanding performance
5. a swivel connector facilitate the removal of 4 water spray fiber optic dental handpiece for daily lubrication.
1. Clean Fiber-Optic Surfaces: Alcohol and a Q-tip may be used to remove excess oil and debris from all fiber-optic surfaces, which will prevent buildup and discoloration.
2. Bag 4 Water Spray Fiber Optic Dental Handpiece: Recommend bagging all instruments and handpieces to maintain sterility. A paper/plastic combination bag allows steam to evaporate through the paper side while the contents remain readily identifiable.
3. Sterilize: Autoclave the 4 water spray fiber optic dental handpiece following the manufacturer's guidelines. Be sure to place the bag paper side up in the autoclave. Allow the sterilizer to completely process through the dry cycle. This step is CRITICAL. If still damp after completion of the full sterilization cycle, leave the bagged instruments in the autoclave until the packaging is dry.
1. Air rotates the turbine at the head (front) of the 4 water spray fiber optic dental handpiece.The turbine consists of two miniature bearings, an impeller and a spindle/chuck assembly for fixing the drill.
2. Air is introduced into the handpiece through the foot pedal of the chair unit. The air flow connected to the back end of the handpiece passes through the handpiece casing and then enters the head of the handpiece. The force is forced through the turbine wheel at the head of the handpiece, causing the turbine to rotate.
3. After the air moves around the turbine wheel, it is forced out of the head and moves downwards through the housing of the 4 water spray fiber optic dental handpiece and through the rear exhaust port of the handpiece.
1 year free warranty for the handpiece, 6 month for the cartridge and 3 month for the bearing.
Zhengzhou Huaer Electro-Optics Technology Co., Ltd. is established in 2006. Located in Zhengzhou city, we're the leading manufacturer for the Portable Dental Unit, Dental Handpiece Teeth Whitening Material and Dental Impression Putty etc. Our products sell well in Australia, Columbia, Kuwait, the Netherlands, South Africa, Thailand, United Kingdom, United Arab Emirates, United States, Zimbabwe. Except our own products, we can also provide OEM services and accept customized order as well. Our products are CE and SFDA certified. For more details, please feel free to contact us to learn more.
Your enquiry will be replied within 24 hours !