From the user's point of view, Dr.Super LED fiber dental handpiece adopts an ergonomic handle and a stable 20000 Lux lighting power drive structure design to meet the needs of medical staff and provide professional treatment for patients.
----With quick-connect design, a key plug, easy installation and disinfection. The user's experience can not be better.
----Bright light broadens the work horizon in the mouth and helps doctors more easily observe the patient's condition
----Lightweight, comfortable, ergonomic handle makes medical staff easier to work
----Push button design for doctors'convenience, allowing doctors to easily change drills
----Quick-connector bulbs for easy installation, an extra bulb accessory for extended life
Operation Air Pressure
≥350(krpm)for Standard head; ≥280 (krpm) for torque head
0.2Mpa ( 2Kgf)
LED Luminous Intesity
1. Closed cartridge protect of bearing and extend lifespan
2. Anti-retraction function design, protect patients from cross infection, extend lifespan for the handpiece
3. 135℃ autoclavable for the handpiece
4. Imported bearings, excellent internal quality and structure, resulting in outstanding performance
5. a swivel connector facilitate the removal of the handpiece for daily lubrication.
1.Improper operation----When doing dental preparation, pressurize the burs endlessly, causing bearing wear, intermittent grinding techniques should be used.
2.The choice of burs----The bur shank of the turbine bur must be standardized. If the bur shank is too thin, sliding between the bur shank and the shaft sleeve can occur, which can increase the wear of the shaft sleeve, resulting in loose cradle pin or flying bur. The bur handle is too thick and damages the clamp shaft when loading and unloading the bur. Also need to promptly replace old boring burs that have been de-sanded
3.handpiece falls, deformation----In the daily work, the handpiece is not placed so well that it may fall or be crashed by the patient. The handpiece will deform and affect the service life.
1.Air rotates the turbine at the head (front) of the handpiece
The turbine consists of two miniature bearings, an impeller and a spindle/chuck assembly for fixing the drill.
2.Air is introduced into the handpiece through the foot pedal of the chair unit. The air flow connected to the back end of the handpiece passes through the handpiece casing and then enters the head of the handpiece. The force is forced through the turbine wheel at the head of the handpiece, causing the turbine to rotate.
3.After the air moves around the turbine wheel, it is forced out of the head and moves downwards through the housing of the handpiece and through the rear exhaust port of the handpiece.
We guaranty each handpiece will be tested strictly before ship to clients. And we promise to repair each handpiece if it's faulty.
Zhengzhou Huaer Electro-Optics Technology Co., Ltd. is established in 2006. Located in Zhengzhou city, we're the leading manufacturer for the Portable Dental Unit, Dental Handpiece Teeth Whitening Material and Dental Impression Putty etc. Our products sell well in Australia, Columbia, Kuwait, the Netherlands, South Africa, Thailand, United Kingdom, United Arab Emirates, United States, Zimbabwe. Except our own products, we can also provide OEM services and accept customized order as well. Our products are CE and SFDA certified. For more details, please feel free to contact us to learn more.
Your enquiry will be replied within 24 hours !